Product group: resealable
How do I clean my resealable flow cell's / OOC top and bottom layers?
Cleaning Resealable flowcells are produced in cleanroom facilities and clean areas to prevent physical impurities. After use, cleaning might be required to remove organic residues. Please use distinguished cleaning processes for glass slides with and without a gasket. Glass slides with gasket: Can be cleaned with ethanol, isopropanol and acetone. The glass slides could be either immersed in the solvents or rinsed with them. Immersion might lead to diffusion of solvents into the gasket. Therefore, it is recommended to rinse the glass slide with the gasket thoroughly with water or buffer to remove residues. Only wipe with lint free tissue, otherwise the sealing might be affected negatively or completely lost. It is advised to not wipe the gasket as this might damage it and reduce its sealing abilities. Physical cleaning of the surface by either heat (with a maximum temperature of 140 °C) or oxygen plasma is possible. Cleaning with acids or bases has not been fully tested. Glass...
How do I sterilize my resealable flow cell's / OOC top and bottom layers?
Sterilization Resealable flowcells are not sterilized when delivered. Please use distinguished sterilization processes for glass slides with and without a gasket. Glass slides with gasket: Can be sterilized with ethanol. Dry heat of up to 140 °C can be applied. Can be autoclaved. Glass slides without gasket: Can be sterilized with ethanol. Can be autoclaved.
What is the oxygen permeability data for the elastomer used?
The oxygen permeability data from the exact material grade used is not available, however the data from a different grade of the same material family, can be found below.