- EUR - Euro
- USD - US Dollar
Microfluidic chips with two channels crossing each other. Also suitable for capillary electrophoresis. The FC_X3550CH.3 has a thin 145 micrometer bottom plate making it suitable for confocal microscopy. Pack of 3 chips.We recommend usage of the Fluidic Connect Pro chip holder for this products. The provided metal strips should be used for this thin bottom product.
|Number of chips per pack||3|
|Distance between channel and top surface||1080 µm|
|Distance between channel and bottom surface||145 µm|
|Total chip thickness||1245 µm|
|Chip size||45 mm x 15 mm|
|Channel width||50 µm|
|Channel height||20 µm|
|Internal Volume||0.1 µl|
|Number of Inlets||3|
|Number of outlets||1|
|Inlet/outlet hole sizes on top of the chip||1.70 mm|
|Inlet/outlet holes size at channel||0.60 mm|
|Optical properties||Optical clear view from all sides|
|Supplied in Fluidic slide?||Yes|
|Material chip||Borosilicate glass|
|Material black cartridge||Polypropylene|
One simple but very effective way to clean a microchip is to flush an alkaline solution through the channels. A solution of 1 M sodium hydroxide in water works well but a lower concentration might also be sufficient. If traces of the cleaning solution remaining inside the chip after cleaning and rinsing with water pose a problem then ammonia can be used instead. Note that these solutions are caustic and can cause damage to e.g. the polyimide coating of fused silica capillaries. Also plastic parts should not be exposed to very alkaline solutions.
In order to aid in the removal of particulate matter, a water bath with ultrasonic agitation can be used, preferably while flushing a watery solution through the channels using a Fluidic connect kit.
Glass microchips can be heated (e.g. >400° C) causing any organic material adsorbed on the glass surface to degrade. Try to use lower temperatures first because burning the content could make it stick. Make sure you only heat the glass chip and not the plastic parts around it.
Concentrated sulphuric acid works well to dissolve organic material such as fibres which are difficult to remove with alkaline solutions, but because of the extremely corrosive nature of the material a cleaning procedure is not so easily implemented.
Please note that chips that were coated by Micronit have different guidelines for cleaning.
Have a look at our clogging prevention guide